Northern Sonoma County Air Pollution Control District 150
Northern Sonoma County Air Pollution Control District 150 Matheson Street Healdsburg, California 95448 (707) 433-5911 George Erdman Sr. Air Quality Specialist III Jessica DePrimo Air Quality Specialist III [email protected][email protected] Data Acquisition & Management Systems: A Small Districts Perspective
(or how I became a kid again at 60) NSCAPCD Monitoring Network 3 PM10 Aerometric Sites w/ MetOne BAM1020-965s and Agilaire ESC 8832s 1 Ozone Aerometric Site w/ Redundant Teledyne API 400E Ozone Analyzers and Agilaire ESC 8832s 3 DSL Lines w/ CISCO VPN Firewalls 1 Satellite Link w/ CISCO VPN Firewall Agilaire AirVision Software on a Virtual Server w/ Separate SQL Database Agilaire AirVision Data Acquisition and Mgt Software AirVision Suite (Version 2.4.161) with: BAM Direct Poll Option (Digital) File Import Configuration Option ADVP (Advanced Data Validation Program) Option Parameter Equation Math Pack Option
Precision & Accuracy Reports Option Why Sample Digitally? Avoid DAC/ADC Conversion Errors Data Reliability Enhanced Instrument Diagnostic Information Available Digital Parameters for API 400E Ozone Analyzer Analog vs. Digital Use of DMS Data in Instrument Diagnostics API 400E Daily Zero-Span Drift and O3 Gen Table Cal Points Daily Ozone Calibration Email
Graphic Representation of May 7 Zero-Span Note lack of feedback response on ozone concentration Normal Daily Zero-Span Note beginning concentration level and end concentration level Real Time Analyzer Manipulation Minute Time Series Graph of O3 Gen Drive (mV), O3 [ppb], and O3 Gen Ref(mV) from API 400E O3 Gen Cal Data Showing APICOM connected to analyzer for control & Hyperterm open on Real-time Data Display O3 Gen Cal Graph from May 8 Note O3 Gen internal calibration points relative to ambient levels required for precision checks Digital Data String for BAM 1020
>4 - Display CSV Data Station, 3 Time, Conc(ug/m3),Qtot(m3),WS(MPH),WD(DEG),BP(mmHg),RH(%),RH( %),AT(C),E,U,M,I,L,R,N,F,P,D,C,T, 05/09/12 08:00, 16, 0.834, 3.17,136.48, 753.5, 19, 56, 19.5,0,0,0,0,0,0,0,0,0,0,0,0, x Data Column Number Data Column Number 10 11 , 0.834 , I
See Comment L See Comment R See Comment CHAN 5 User Edit CHAN 6 User Edit (WD) (Deg)
BAM Data File Import Template: Error Code Parsing Advanced Data Validation Processor Task: Poll Raw Data from Sites AV Database Task: Apply ADVP Rules Raw Data Flags Null Codes Email Notifications ADVP Editor BAM Email Error Messages
From: [email protected] To: George Erdman; Jessica DePrimo; Randall Woodward; Alex Saschin Subject: BAM Error Flag "D" - BAMFLAG_D 5/1/2012 12:00:00 - 5/1/2012 13:00:00 1h Date: Tuesday, May 01, 2012 2:04:37 PM Importance: High Message: The BAM Station below has recorded an Error Flag "D" The error indicates that the reference membrane span check measurement (m) for that hour was out of agreement with the expected value (ABS) by more than 5%. Applying Actions on reading: Healdsburg:BAM01_PM 10:001h at 5/1/2012 12:00:00 with value 8 ------ > Annotation Added "Ref Membrane BAM CAL Error; does NOT invalidate hourly data" ------ > Add Flag ?
From: [email protected] To: George Erdman; Jessica DePrimo; Randall Woodward; Alex Saschin Subject: BAM Error Flag "L" - BAMFLAG_L 5/1/2012 13:00:00 - 5/1/2012 14:00:00 1h Date: Tuesday, May 01, 2012 3:04:33 PM Importance: High Message: The BAM Station below has recorded an Error Flag "L" The error occurs any time power is
cycled or lost, even momentarily. See manual Section 6.5. Applying Actions on reading: Healdsburg:BAM01_PM 10:001h at 5/1/2012 13:00:00 with value 985 ------ > Annotation Added "Power Failure Error" ------ > Apply Null Code AV ------ > Add Flag P Daily Parameter Report Benefits of Enhanced DAS/DMS 50% Reduction in Trips to Sites Estimated 60% Reduction in Data Review and Submittal Shortened discovery time equipment failures Reduced data loss from failures Enhanced Diagnostic Abilities Remote Operation of Equipment
Jenna Warren, Counselor. Additional Support Members: Katie Remis, Psychologist. Raelene Daley, Counselor. WCPSS Vision of MTSS. Asystemic multi-tiered instructional framework that guides instruction and intervention that is matched to student need.
RI training for HDR students is located on your Blackboard account under 'My Units' All HDR students must complete this training before Candidacy. There is an assessment, pass grade is 80%. The RI includes 'Originality Checking' links to Turnitin (text...
In the Jolly Phonics group the 'tail' of low achievement in writing was simply abolished. This claim is supported by research; Jolly Phonics has been independently studied in the UK and overseas with results showing it is an incredibly effective...
May be carried forward as an asset in certain circumstances (e.g. when labour cost relates to inventories - work in progress) (AASB 102, para 12) Cost of employee benefits can also be included in cost of property, plant and equipment...
At Burnaby Central Secondary . . . Students who participate in the following programs at Central do NOT need to fill out DPA logs for the year: AVPA PASS Whitecaps Pre-Residency Girls Soccer Program G-REX BWC Hockey Academy At Burnaby...
Biomes. Extends along the Atlantic coast of Brazil and inland as far as Paraguay and Argentina. Atlantic Forest. Although it is just a small fraction of the size of the g Amazon rainforest, the Atlantic Forest still harbors of a...
- Legal Advice - Business Services - Indigenous Policy Lobbying - Local Commercial Generating. Indigenous Services Pty Ltd is the Trustee of the Nyamals People's Trust. They are particularly keen to assist the Nyamal to improve their well=beingand not just...
Ready to download the document? Go ahead and hit continue!